Includes the latest and most frequently asked questions regarding protocol products covering Bluetooth, DDR3 & DDR4, Ethernet, Fibre Channel, IEEE1394, Infiniband, PCI Express, PCI/PCI-X, SAS, SATA, USB, and others.
When capturing in Recirculate mode, the analyzer does not transfer capture data to the host until the capture is stopped, at which time the program application can then access the capture data and create a trace file on disk.
Note: Current View windows for Phase and Timing will remain blank until such time as capture is stopped. Their presence is to permit these views to be "kept" for the display of the capture data after capture has been stopped. Also note that when there is no capture activity there will be a message on the status bar indicating the length of time the capture activity has been quiescent in seconds.
(All support files listed here have been scanned and are virus free)